From: Jason Curtis - University of Florida
Subject: Re: [ISOGEOCHEM] Clearing blocked Valco valve in Gasbench
Date: September 9, 2009 10:00:48 AM PDT
We also had a clog in our valco in our gasbench. The valco valves are actually really easy to take apart and almost as simple to rebuild. I am 99% sure that the instructions to do this are on the valco site. Basically you remove the top nut without changing the set screw. Then cycle the valve while holding a magnet on the top of the rotor (I used a strong magnet from inside a broken hard drive, they suggest a pencil type magnet). This allows the "surface tension" to be broken and allows you to lift the rotor out. Then you can see the holes for each port. Shine a light in each port and see if you see light from the top. I blew out the junk that was blocking ours. You might have to use water. Whatever you do be careful not to scratch the inside of the valve - if this happens it must go back to valco to be reground. When putting it back together there is a mark on the rotor that goes to the correct port - see the manual as it is different for each type of valve.
A note on rotors. When we scratched a rotor I called valco to get a new one. They did not have the rotors in the web catalog in stock and it was going to be weeks. After trying to figure out how to get a new one short of buying an entire new valve the person from valco suggested that they have different rotors made of different materials. They are not in the catalog. Different rotors of different material are for different temperature ranges. The one that came in my gasbench was good to like 450°C. The one that I got was good to like 350°C. A little hotter than we need in the gasbench which sits at like 22°C!
Hope this helps. Let me know if you need further info. Jason
Making Bench Sample Needle
Jonathan Wynn wrote:
Has anyone out there successfully replaced the fused silica capillary in double-holed needles used by the Thermo Finnigan Gasbench? Suggestions or tips would be very welcome on how to extract the fused silica capillary from the needle's interior (what sort of glue is used?), and rethreading a replacement piece.
Rather than treat these as consumables at ~$400/ea., it would be great if there were a simple method of replacing the capillary, which is not only susceptible to clogging (relatively easily solved), but the occasional breakage.
Department of Geology
University of South Florida
We've been making our own for a while now.
A 19Ga needle, ~4" (10cm), with a hole in the side, ~ 1cm up from the tip, does the trick. We epoxy it in place--but there was mention on Isogeochem that if the hole is far enough away from the tip, you don't need to do that.
To drill the hole in the side yourself, clamp the needle between two boards, with a small groove in one to hold it in place and a guide hole in the other for the (dental) drill. It may take a little practice. You can buy a custom made needle--we got some from Popper and Sons in the states--but that only make sense if you buy in bulk.
The needle attaches to a standard 1/16" swagelok 3-way connector (or metric equivalent). The glass capillary feeds straight through the connector and down the needle. We still epoxy it in place at the bottom of the needle--push extra through then snap that off when the epoxy dries. You'll want a tight fit on the needle. We like to use a stainless steel connector since you have have to over tighten the standard 1/16" ferrules to get them to grip the needle. We'll use the Valco ferrules (longer) in the swagelok fitting, or sometimes use a brass swagelok ferrule with an extra backing ferrule. The capillary ferrules are the standard GC capillary ferrules.
For attaching to the PAL holder, we use a 1/16" to 1/8" swagelok reducing union. The key connection is the attachment to the needle. We use a metal ferrule to tightly hold the needle in place--usually by pre-swaging a steel valco ferrule in a valco union then sliding onto the needle, or even getting it swaged (is that a word?) onto the needle before starting anything else. On the 1/8" side, we'll use an 1/8" to 1/16" reducing ferrule as a guide only. Please note, when attached the needle tip should not be below the bottom of the PAL's bottom needle guide.
The side arm is for the He feed. We're still using the steel capillary originally from Thermo--but some are available from McMaster-Carr (cleaning needed?).
We make homemade acid needles by just sharpened the stainless steel capillary (from McMaster-Carr in the US). In this case were using a 1/16" to 1/8" adaptor to fit into the 1/8" swagelok union--use the parts you got...
Question: May I know how could you solve the problem of clogging of the needle?
Answer: We place the needle under a binocular microscope and clean out septum crumbs with a fine wire. Keeping the needle sharp with a file helps extend the life greatly.
SIL Co-Director Matt McCarthy directed a
weekend program for about 40 Educational Opportunity (EOP) STEM transfer students highlighting the use of stable isotope analyses in phyiscial and biological science inquiry.
SIL Co-director and UCSC Physical and Biological Sciences Dean Paul Koch has been elected 2015 AAAS Fellow for "energetic and innovative leadership in applying stable isotope geochemistry to document and interpret environmental change through the past 65 million years of Earth history.".
Formation of coastal sea ice in North Pacific drives ocean circulation and climate. So indicates evidence published by UCSC scientists Karla Knudson and Christina Ravelo.
Christina Ravelo, Professor of Ocean Sciences at UC Santa Cruz and UCSC Stable Isotope Laboratory co-director, has received the 2013-14 Outstanding
Faculty Award from the Division of Physical and Biological Sciences. The annual award is the division's highest
honor for faculty achievement, recognizing combined excellence in research, teaching, and service.
Long-term human influenced dietary habits of Yosemite National Park black bears explored by UCSC scientist Jack Hopkins, Paul Koch, and colleagues.
SIL Co-Director Paul Koch honored as a Fellow of the California Academy of Sciences. See story here.
Diversity of great white shark diet was shown to be greater than previously thought based on research by UC Santa Cruz colleagues Sora Kim, Paul Koch, and James Estes along with co-author M. Tim Tinker of the U.S. Geological Survey. This work was highlighted in the Los Angeles Times.
SIL Co-Director Paul Koch has been named Dean of the UC Santa Cruz Division of Physical and Biological Sciences.
SIL Co-Director Paul Koch was quoted in the New York Times on research showing that a tiny Paleocene horse Sifrhippus, the first horse, grew even smaller during the Paleocene-Eocene Thermal Maximum climatic warming event.
SIL Co-Director Christina Ravelo has been selected as an American Geophysical Union (AGU) Fellow.
SIL Co-Directors Paul Koch and Jim Zachos's research on Paleocene-Eocene Thermal Maximum (PETM) climatic warming event is featured in an National Geographic article "Earth Before the Ice" in the October 2011 issue.
SIL Co-Director Paul Koch has been appointed Interim Dean of the UC Santa Cruz Division of Physical and Biological Sciences.
SIL co-director Christina Ravelo (Ocean Sciences) sails as co-chief scientist on Integrated Ocean Drilling Project (IODP) Expedition 323 to investigate Bearing Sea Paleoceanography.
SIL co-director Christina Ravelo (Ocean Sciences) gives the 2008 Emiliani Lecture at the American Geophysical Union Meeting in San Francisco on "Lessons from the Pliocene Warm Period and the Onset of Northern Hemisphere Glaciation".
UCSC SIL has been funded by the National Science Foundation for a new Dual-Inlet Isotope Ratio Mass Spectrometer with individual acid drip system for very small calcium carbonate samples.
SIL co-director Jim Zachos (Earth and Planetary Sciences) recieves prestigious Humbolt Research Award. See Humbolt Award for details.
SIL Co-Director Jim Zachos has been selected as an American Geophysical Union (AGU) Fellow.
May 19, 2006: UCSC Stable Isotope Laboratory Symposium
In recognition of the new continuous flow instruments added to the UCSC stable isotope facility a Symposium is being run to highlight the new analytical capabilities.