Useful references for Elemental Analyzer samples
Blank correction methodology for EA δ13C and δ15N data
Brian Fry, Willi Brand, F. J. Mersch, K. Tholke, R. Garritt, Automated analysis system for coupled &delta13C and δ15N measurements , Anal. Chem., 1992, 64 (3), pp 288291, DOI: 10.1021/ac00027a009
Sulfur Isotope Analysis techniques.
Brian Fry, Steven R. Silva, Carol Kendall, Richard K. Anderson, Oxygen isotope corrections for online δ34S analysis, Rapid Communications in Mass Spectrometry, v.16, p.854-858.
Siderite removal for δ13C, %C and C/N ratio analysis.
Larson, T.E., Heikoop, J.M., Perkins,G. Chipera, S., Hess, M. (2008) Pretreatment technique for siderite removal from organic samples for carbon isotope and C:N analysis in geological samples., Rapid Communications in Mass Spectrometry, v.22, p.865-872.
Macrofauna calcification removal (e.g., Sea Urchins).
Kolasinski J, Rogers K, Frouin P 2008, Effects of acidification on carbon and nitrogen stable isotopes of benthic macrofauna from a tropical coral reef. Rapid Commun. Mass Spectrom. 2008; 22: 2955-2960
Choice of Solvents on δ13C EA results.
Schlechtriem, C. H., Focken, U. & Becker, K. (2003). , Effect of different lipid extraction methods on d13C of lipid and lipid-free fractions of fish and different fish feeds. Isotopes in Environmental and Health Studies 39, 135-140.
Ruiz, N., Dubois, N., Wielgosz-Collin, G., du Pont, T. R., Berge, J.-.P, ouchus, Y. F., Barnathan., G. (2007). , Lipid content and fatty acid composition of a marine-derived Trichoderma longibrachiatum strain cultured by agar surface and submerged fermentations. Process Biochemistry 42, 676-680
Delta+XP 24VDC/5VDC 3rd Party Power Supply
Supplier: Integrated Power Designs: www.ipd.power.com Phone: 570-824-4666
Dual output power supply. Product no. CE-225-2101 with Cover - $273.60
Though size wise identical it is not just a plain swap. You have to tweak it a bit.
- To mount it on the metal plate (the original supply was on) you have
to drill holes on the back plate by removing the board.
I got all these from mouserelectronics.com ( http://www.mouser.com/catalog/catalogUSD/641/1460.pdf)
The part numbers are: 538-09-50-3031 (three pin)
- The original specs of the Philips power supply are as follows: AC 110/230V, 6A/3A 47-63Hz DC 24V/8A DC5V/6A
- When using CE-225-2101 or any other compatible power supply, there is no need to re-connect the jumpers as on the original. Just the Brown and blue leads for 24V and 5V, that's all.
Conflo III and Delta Plus XP Manuals and Brochures